The TR Schedule Pod is used for scheduled long term recording using up to 12 Telemetry Research transmitters in conjunction with one receiver. The TR Scheduler Pod switches the receiver to record from multiple individual transmitters through a single PowerLab input. The pod is controlled using the supplied TR Scheduler Extension for LabChart. The pod and extension allows data recording durations and periods to be specified allowing a system to record from hours to days from multiple telemetry transmitters. A User?s Guide containing more detailed information is installed with the Extension and is also available in the Help menu of LabChart. System Compatibility: The ML319 TR Scheduler Pod connects to any PowerLab hardware with Pod ports (8-pin DIN inputs) and requires the TR Scheduler LabChart Extension to operate. The TR Scheduler Pod is supported by the following versions of LabChart and Scope software: WINDOWS ? LabChart v3.4.8 or later ? Scope v3.6.3 or later MACINTOSH ? LabChart v3.6.3 or later ? Scope v3.6.3 or later Note: Earlier software versions do not support Pods. TR Scheduler Pod Capability: The following items are suitable for use with the TR Scheduler Pod: ? MLE1051 Wireless Biopotential/Temperature Receiver ? MLE1053 Wireless SNA/Temperature Receiver Stacking and Unstacking Pods: Pods stack by clicking into place on top of each other. To separate stacked Pods, push the top Pod towards the back and then pull them apart from the back.
|Cable Length Metric||DB15: 1.4 m; DIN: 1.4 m|
|Enclosure Size L x W x H Metric||110 x 60 x 35 mm|
|Operating Conditions||0 ? 35 °C, 0 ? 90 % humidity (non-condensing)|
|Output Connection Type||DB15 to Telemetry Research receiver DIN to PowerLab pod input BNC (reserved)|
|Product Family||Data Acquisition|
|Weight Metric||~200 g|