IHSR-WH-LVPM LVP Measurement in Mouse Working Heart, 73-4037
The IH-Series includes a specialized pathway which easily allows introduction of a Tip Pressure Catheter 0.8F (LVP) directly into the left ventricle via the adapter port and the aorta, rather than via apical puncture. The Tip pressure Catheter is connected using a dedicated connecting cable to a TAM-A Tranducer amplifier module
|Analog Indication||HEC-10D||TAM-A LED bar graph 20 LEDs (+13/-7) for visual check on the signal sensitivity approx. 1V/LED TAM-D 3 1/2-digit LED display|
|Bridge Balance||PLUGSYS||Through 10-turn trimmer coarse adjustment and electronic autozero by push button (LED for error if autozero is not possible)|
|Bridge Supply Voltage||+5V / 50 mA max.|
|Electrical Calibration||Selectable by switch on front panel:
a) 0 V output signal with switch in position '0'
b) Positive or negative calibration output voltage adjustable with 10-turn trimmer if switch is in position 'CAL'
|Gain||Selectable ranges by internal Jumper: 0.2 to 10, 0.4 to 20, 1 to 50, 2 to 100, 4 to 200, 10 to 500, 20 to 1000, 100 to 5000, 200 to 10000. Fine Adjustment Through 10-Turn Trimmer|
|Output Low-Pass Filter||a) Selectable by switch on front panel for pulsatile output signal: 1, 100, 300 Hz
b) Selectable by internal jumper for mean output signal: 0.1, 0.3 Hz
|PLUGSYS Width*||2 slot units|
|Power Supply||+5V through connector from PLUGSYS bus system|
|Signal Output||a) On front panel through BNC socket|
|Transducer Input||6-pin socket with screw lock (binder, Amphenol Tuchel) Differential input circuit, imput impedance 1010?|